Screen: 10.4 '' LCD Display Screen
Suitable Adult & Child
Mode: pneumatically powered and electrically controlled system
Working Mode: Closed; Semi-Closed; Semi-Open
Circuit Breathing circuit integrated standards
Flowmeters: 5 Tubes Flowmeters: O2:0.1~10L/Min, N2O:0.1~10L/Min; Air: 0.1~10L / Min
Trolley: Fitted with 4 nos anti-static rubber castors; two of which are lockable for braking and easy maneuverability with foot operated brake provisions
Drawer unit TWO drawers totally extractable
Gas requirement: Medical oxygen and nitrous oxide with a pressure ranging from O2: 0.32~0.6MPa ; NO2: 0.32 MPa to 0.6 MPa.
safety valve <12.5 kPa
Respiratory Rate 1~99bpm
oxygen concentration in the mixed gas N2O/O2 > 21%
Oxygen Flush: 25~75 L/min
modes of ventilation A/C, IPPV, SIPPV, IMV, SIMV, PCV, VCV, PEEP, MANUAL, SIGH
PEEP: 0 ~ 2.0 kPa
Inspiratory trigger pressure -1.0kPa ~ 2.0 kPa
IMV frequency: ..
I/E Ratio: 8:1 ~ 1:10, Has the inverse ratio ventilation
Tidal Volume 0 ~ 1500 ML
Inspiratory Plateau: 0~1s
O2 Concentration: 21%~100%
Sigh: one deep breath per 70 ~ 120 controlled respirations, the inspiration time is 1.5 times of the setting point
Maximum safety pressure: <= 12.5 kPa
Pressure limit range: 0 ~ 6.0 kPa
Airway pressure alarm: Audible and visual and with yellow and red color indicating Lower: 0.2kPa~5.0kPa; Upper: 0.3~6.0 kPa
Tidal volume alarm: Audible and visual and with yellow and red color indicating upper alarm: 50 to 2000ml, lower alarm: 0~1800ml
Oxygen concentration alarm: Audible and visual and with yellow and red color indicating upper alarm: 21% ~100%; lower alarm:10%~80%
Power Supply Alarm "Ac/dc power supply are after failing to send out alarm immediately
Alarm Time: keep >120s"
The airway pressure continues to be higher than 15 hPa?±1 hPa for 15s±1s, then the machine will raise an audible alarm, the pressure will be displayed in red and the continuous high pressure red alarm wording is displayed on the screen of the anesthetic respirator.
Ambient temperature: 10 ~ 40oC
Relative humidity: no higher than 80 %
Atmospheric pressure: 860 hPa ~ 1060 hPa
Power requirement: 100-120 Vac, 50/60 Hz;
Attention: the AC power supply used for the anesthesia machine must be well grounded.
Attention: the anesthesia machine used must be equipped with a carbon dioxide monitor complying with ISO 9918:1993, an oxygen monitor complying with ISO 7767:1997 and an expiratory gas volume monitor complying with 220.127.116.11 of Medical Electrical Equipment Part II: Special Requirements for Safety and Basic Performance of Anesthesia System.
Ambient temperature: -15oC?~ +50oC
Relative humidity: no higher than 95 %
Atmospheric pressure: 86 kPa ~ 106 kPa.
It should be stored in a room without corrosive gas and well ventilated
packaging box comply with the requirement of GB/T 15464
Between the packaging box and product, soft material with appropriate thickness provided to prevent loosening and mutual friction during transportation
Moisture protection and rain protection to ensure that product is protected from natural damage.
Safty & Alarm
Oxygen Alarm It alarms when the oxygen supply from pipe or cylinders lower than 0.2MPa
Ventilation Volume Alarm Lower: 0~12L/Min; Up: 18L/ Min
Power Alarm It Alrmas while AC and DC supply failure ; Keep alraming time: >120s
Air Tract Pressure Alarm Lower: 0.2kPa ~5.0 kPa; Up: 0.3kPa ~ 6.0kPa